You have no obligation to purchase the product once you know the price. You can simply remove the item from your cart. Test Equipment Test Equipment has always been proud of being one of the most progressive specialist suppliers of test and measurement equipment. Australian owned and operated by experienced engineers, we offer extensive product and application knowledge as well as quality service, which translates into a seamless experience when you procure your test and measurement instruments from us. We started as a small business, fueled by the knowledge that we could make a positive impact and have come a long way since our inception, growing a small range of products into a full range of hundreds of products from some of the most respected brands in the industry such as AEMC, FLIR Systems, Kyoritsu, Megger and Testrix Systems.
|Published (Last):||13 August 2015|
|PDF File Size:||1.45 Mb|
|ePub File Size:||16.70 Mb|
|Price:||Free* [*Free Regsitration Required]|
KYORITSU will not accept any liability for any damage or injury caused by misuse or non- compliance with the instructions or safety procedures. Return the instrument to Kyoritsu or your distributor for service and repair to ensure the safety features are maintained. Page 6: Features 2. Features MODELA is an analogue tester with five ranges for insulation resistance measurement and continuity testing resistance tests of low voltage installations.
Page 7: Specifications 3. Page Instrument Layout 4. Preparation for Testing Mechanical Zero Adjustment Check that the pointer lines up with the middle of the mark on the scale correctly. If not, adjust it by rotating the meter movement zero adjust with a screwdriver, etc. This may damage the circuit. Voltage check can be made with the range selector switch at any position.
Connect the test probe to the circuit under test and read a voltage value. If the circuit is live, the meter indicates the voltage, the live circuit lamp is lit, and warning buzzer sounds. Page Continuity Testing resistance Tests required test voltage when providing a steady state current of 1mA.
The minimum allowable resistance level is 0. To avoid possible electrical shock, disconnect the test probe before opening the cover for battery and fuse replacement. Page Notes On Accessories 8. Notes on Accessories Case Lid The case can be fitted under the housing case as illustrated bellow. Page Cleaning Of The Instrument 9. But in the dry time of winter static electricity sometimes builds up on the meter cover due to the characteristic of plastic.
เครื่องวัดความต้านทานลูป LOOP/PSC/RCD TESTER
Kyoritsu 5406A RCD Tester